Please login or register. Did you miss your activation email? This topic This board Entire forum Google Bing. Print Search. Pages: [ 1 ] Go Down. I was able to install the drivers fine, but when I run the PC software setup, it pops up an error dialog that would probably be in Chinese if I had the right fonts installed. And that's all it does. It is happening both on my Dell laptop with Windows 10 and my Desktop with Windows Anybody here have any experience or ideas?
Mine installed correctly on Windows 10 but I used a package downloaded from Owon, not the CD that came with it. Maybe there was a bug on the version that comes with CD in box?
You can always learn Chinese though. There are only over characters you'll have to learn I've not tried Win Is there a possibility that you are installing under an account without sufficient permissions IIRC it says somewhere in their docs that you need an account with administrator privileges. The scope would disconnect 10 seconds after going idle.
I verified this problem with 4 separate PC systems, using multiple versions of the software. Star This commit does not belong to any branch on this repository, and may belong to a fork outside of the repository. Branches Tags. Could not load branches. Could not load tags. A tag already exists with the provided branch name.
Many Git commands accept both tag and branch names, so creating this branch may cause unexpected behavior. Are you sure you want to create this branch? Local Codespaces. Sign In Required Please sign in to use Codespaces.
Launching Xcode If nothing happens, download Xcode and try again. Launching Visual Studio Code Your codespace will open once ready. Latest commit. Git stats 19 commits. Failed to load latest commit information. October 23, December 26, July 24, April 17, February 9, View code. Linux Debian based, Arch based, Puppy, Fedora If you wish to calibrate the device manually then: Disconnect the probes Press F2 to open the calibration dialog Select the Zero compensation tab For each voltage, adjust the calibration as to obtain a 0v signal Select the Zero amplitude tab For each voltage, adjust the calibration as to obtain a 0v signal Select the Gain tab For each voltage, connect the probe to a reference voltage and adjust the calibration The factory calibration is stored directly in the device in the flash memory.
Releases 15 tags. You signed in with another tab or window. Reload to refresh your session. You signed out in another tab or window.
Learn more. Please sign in to use Codespaces. If nothing happens, download GitHub Desktop and try again. If nothing happens, download Xcode and try again. There was a problem preparing your codespace, please try again. Java Runtime Environment 8 1. Java 11 or superior is recommended especially if you have an HDPI display. The script installs the driver, copies the files, registers for uninstall and creates a menu entry.
Open a terminal window in this folder and execute sudo bash install-linux. The script builds a package according to the distribution and installs it with the default package manager. To debug, run owon-vds-tiny from a console. Open a terminal window in this folder without Rosetta and execute sudo bash install-mac. The factory calibration is stored directly in the device in the flash memory.
The current calibration is stored in the user folder as "VDSxxxxxx-cals. This project provides a Python API to directly communicate with the device. It can be used for data logging or to analyse and visualise the samples in a Jupyter Notebook. This device supports the SCPI protocol once the application is lauched and running. The server is off by default. Go to Utility to enable it.
This scope can measure 40v peak to peak with a 1x probe and v with a 10x probe. To safely probe between two potentials, even if it's a Neutral, either use two probes and substract the waves Math menu or use a differential probe.
You shouldn't connect the ground of the probe to a potential unless you know exacly what you are doing. It should be the case with the isolated version VDSi. Be also aware that the channels have a common ground which would create a short if connected to two different voltages. If you want to know more on the subject, search for "ground loop oscilloscope" and "earthing system".
Skip to content. Star I verified this problem with 4 separate PC systems, using multiple versions of the software.
Regards, Dave. Gyro Super Contributor Posts: Country:. Jdterry Newbie Posts: 1. Having the same issue with CD pc software installation blocked with cryptic error method using win 7 64 on several machines. Was unable to download software from Owon site. Has anyone found a way to load the software. I've uploaded my personal archive again Try version 1. No idea what's going on with the Owon site at the moment! I think only trap is you have to install drivers manually in the windows device manager - there will be some unknown device - that is your scope.
If you must use the shortest ones and use both of them in parallel. With shitty extension cables it will not work better option or it will measure nonsense I "measured" over 7V at 5V signal - there is probably problem with voltage reference when there is higher voltage drop at USB cable.
Pages: [ 1 ] Go Up. There was an error while thanking. SMF 2. EEVblog on Youtube.
WebFujian Lilliput Optoelectronics Technology Co.,Ltd Add:No Heming Road Lantian Industrial Zhangzhou P.R. China Tel:+ Fax:+ E . WebOct 23, аи Download and extract the latest release: saadpcsoftware.com Windows 10, 8, 7, XP (x86/amd64) . WebMar 31, аи Using the method of installing the software first, then manually installing the drivers from: C:\Program Files (x86)\OWON\VDS_C2\USBDRV Resolved my disconnect .